HIOKI 3154 MANUAL PDF
View and Download Hioki instruction manual online. DIGITAL M? HiTESTER. Test Equipment pdf manual download. Hioki Pdf User Manuals. View online or download Hioki Instruction Manual. Hioki Digital M-Ohm HiTester, Hioki – on Sale at the Test Equipment Depot. Slow: less than seconds (at manual ranging). Sampling rates, Fast: .
|Published (Last):||13 May 2015|
|PDF File Size:||20.29 Mb|
|ePub File Size:||5.55 Mb|
|Price:||Free* [*Free Regsitration Required]|
After the start of the test, comparator judgment operation can be prohibited until a set interval from 0. Setting Judgment Beep Sound Rapidly assess in as fast as 50 ms Testing is complete in as little as 50 ms hioi the fastest time in the industry.
Powering On and Off the Unit See 2. Test voltage Discharge time Approx. Saving And Loading Test Conditions Blinks during discharging buildup in tested object.
Setting Test Signal Off Timing Multiple query commands can also be in a line. This is ms faster than legacy Hioki models. F when exceeding maximum Setting And Verifying Comparator Connecting Cord and Probes See 2.
For details, see 3. The output queue is also cleared when the power is turned off and turned on again. Indicates that incorrect operation presents an extreme hazard that could result in serious injury or death to DANGER the user.
When interface cable, etc. Queries measurement value and judgment result. When disconnecting from thedisconnect test probe first and signal lead last. LOAD Loads test conditions SW Analog output accuracy: Beeps for PASS judgment. Enter text from nioki Chapter hikki Maintenance And Service Pack the product carefully so that it will not be damaged during shipment, and include a detailed written description of the problem.
Hioki 3154 Instruction Manual
Each setting display blinks during setting status. Communication conditions are not configured properly. Explains the received data. During testing, analog output follows the measurement display timing.
Insulation Tester ST – Hioki
Definition of test duration: RANGe Set analog output to output hioko all measurement resistance ranges or in each range display area. After completing testing, analog output continues to output and hold final voltage. The uses character string data and numeric data, and the type use varies according to the command in question. Changes to Specifications Changes are made to software version 1. High voltage occurs measurement between terminals.
The contact check function allows you to check that proper contact has been made with the device under test prior amnual the testing. Delay time countdown 1. Please select and purchase the optional test leads to accommodate your application. Don’t show me this message again.
Replace the test probes specified by Hioki. Following indicator appears displaying a number blinking to display save maual. Chapter 1 Outline The ST and ST cannot be operated alone. Power is not turned ON all connected equipment. There are two methods: When more than 60 mA current is required, current amplifier transistor circuit and other external power supply devices must be connected externally.
Hioki Digital M-Ohm HiTester, Hioki – at the Test Equipment Depot
Notes On Use Using the product under such conditions could result in electrocution. Do not keep power on equipment to be tested when connecting probes.
All saved test conditions hiooki cleared by system reset. Use the original packing materials when reshipping the product, if possible. Test voltage is V. All signal leads are isolated through a photocoupler from other parts. When timer indicator is blinking, press both key and key at the same manal to display “–” blinking or hold down either key or key to display “–” blinking.
Printing Example After maximum data assigned number is reached, it starts from number 1 again.